Document Server@UHasselt >
Research publications >
Please use this identifier to cite or link to this item:
|Title: ||Distal radius plate of CFR-PEEK has minimal effect compared to titanium plates on bone parameters in high-resolution peripheral quantitative computed tomography: a pilot study|
|Authors: ||de Jong, Joost J. A.|
van Rietbergen, Bert
Arts, Jacobus J.
Geusens, Piet P.
van den Bergh, Joop P. W.
Willems, Paul C.
|Issue Date: ||2017|
|Publisher: ||BIOMED CENTRAL LTD|
|Citation: ||BMC MEDICAL IMAGING, 17, p. 1-7 (Art N° 18)|
|Abstract: ||Background: Carbon-fiber-reinforced poly-ether-ether-ketone (CFR-PEEK) has superior radiolucency compared to other orthopedic implant materials, e.g. titanium or stainless steel, thus allowing metal-artifact-free postoperative monitoring by computed tomography (CT). Recently, high-resolution peripheral quantitative CT (HRpQCT) proved to be a promising technique to monitor the recovery of volumetric bone mineral density (vBMD), micro-architecture and biomechanical parameters in stable conservatively treated distal radius fractures. When using HRpQCT to monitor unstable distal radius fractures that require volar distal radius plating for fixation, radiolucent CFR-PEEK plates may be a better alternative to currently used titanium plates to allow for reliable assessment. In this pilot study, we assessed the effect of a volar distal radius plate made from CFR-PEEK on bone parameters obtained from HRpQCT in comparison to two titanium plates. Methods: Plates were instrumented in separate cadaveric human fore-arms (n = 3). After instrumentation and after removal of the plates duplicate HRpQCT scans were made of the region covered by the plate. HRpQCT images were visually checked for artifacts. vBMD, micro-architectural and biomechanical parameters were calculated, and compared between the uninstrumented and instrumented radii. Results: No visible image artifacts were observed in the CFR-PEEK plate instrumented radius, and errors in bone parameters ranged from -3.2 to 2.6%. In the radii instrumented with the titanium plates, severe image artifacts were observed and errors in bone parameters ranged between -30.2 and 67.0%. Conclusions: We recommend using CFR-PEEK plates in longitudinal in vivo studies that monitor the healing process of unstable distal radius fractures treated operatively by plating or bone graft ingrowth.|
|Notes: ||[de Jong, Joost J. A.; van den Bergh, Joop P. W.] Maastricht Univ, NUTRIM Sch Nutr & Translat Res Metab, Maastricht, Netherlands. [de Jong, Joost J. A.; Geusens, Piet P.; van den Bergh, Joop P. W.] Maastricht Univ, Dept Rheumatol, Med Ctr, Maastricht, Netherlands. [Lataster, Arno] Maastricht Univ, Dept Anat & Embryol, Maastricht, Netherlands. [van Rietbergen, Bert; Arts, Jacobus J.] Eindhoven Univ Technol, Fac Biomed Engn, Eindhoven, Netherlands. [Arts, Jacobus J.; Willems, Paul C.] Maastricht Univ, Dept Orthoped Surg, Med Ctr, Maastricht, Netherlands. [Arts, Jacobus J.; Geusens, Piet P.; Willems, Paul C.] Maastricht Univ, CAPHRI Sch Publ Hlth & Primary Care, Maastricht, Netherlands. [Geusens, Piet P.; van den Bergh, Joop P. W.] Hasselt Univ, Fac Med & Life Sci, Hasselt, Belgium. [van den Bergh, Joop P. W.] VieCuri Med Ctr, Dept Internal Med, Venlo, Netherlands.|
|ISI #: ||000395943900001|
|Type: ||Journal Contribution|
|Validation: ||ecoom, 2018|
|Appears in Collections: ||Research publications|
Files in This Item:
|Published version||799.97 kB||Adobe PDF|
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.